when you encounter some common failures of salt spray environmental chambers, how to solve them? wuxi guanya refrigeration technology co., ltd. has brought the relevant fault tips as follows.
北京幸运飞艇开奖直播 salt spray environmental chamber cannot spray
failure causes: 1. air compressor does not work; 2. main switch of the air compressor outlet does not open; 3. solenoid valve malfunctions; 4. pressure meter malfunctions or the pressure is too low; 5. electromagnetic contactor malfunctions; 6. nozzle blocks;
handling methods: 1. open the air compressor; 2. open the main switch of the air compressor; 3.4.5. ask the after-sales department for door-to-door repair service; 6. remove the nozzle and clean it (please carefully disassemble).
the temperature of the saturation tank of the salt spray environmental chamber cannot rise to the set temperature
failure causes: 1. the temperature of the temperature controller of the saturation tank is too low; 2. the temperature of the saturation bucket safety protection switch is set too low; 3. heating system malfunctions; 4. electromagnetic relay malfunctions; 5. controller malfunctions.
handling methods: 1. set the temperature of the temperature controller to the required temperature; 2. set the temperature of the safety switch to the required temperature; 3. ask the after-sales department for door-to-door repair service.
lack of spray amount in the salt spray environmental chamber
failure causes: 1. spray regulator is placed too low; 2. glass filter pf the preheating tank is blocked; 3. pressure is set too low.
北京幸运飞艇开奖直播 handling methods: 1. raise the spray regulator; 2. clean up the glass filter; 3. adjust the pressure of the pressure valve to 1kg/cm², the air compressor is marked with a pressure valve, adjust to the pressure to 2kg/cm².
These are some handling methods of the salt spray environmental chamber failures brought by Wuxi LNEYA. Wuxi LNEYA has environmental chambers for sale, the quality and the after-sales are guaranteed.